HD-F764 Mattress Roller Durability Tester- EN1975
-Durability Test + Hardness & Height Change Test + Edge Test
for reference only
Cylinder Wood Barrel (EN 1957)
140kg, length 1000m, Diameter 300m
Maximum size of mattress
According to different mattress to adjust the rolling stroke (adjust the limit switch position)
Test Stoke:250mm on either side of the longitudinal centre line
Low coefficient of friction guide, ensure full drum weight on the mattress.
The LCD Touch Screen display provides easy control and setting.
Test Standard: EN 1975
Applications & Test Purpose
The Mattress Durability Tester has been manufactured to determine the durability of any type of mattress and box spring unit, regardless of the materials of construction, by simulating long-term use conditions. Using a cylinder (both ends of the diameter of 250 mm, the middle part of the 300mm), rolling back and forth on the mattress, simulate human motion to assess the durability of mattress.
EN Hardness & Height Change Test
Diameter 355 mm, Spherical radius of curvature 800mm, Front edge radius of curvature 20mm
SMC brand electric cylinder
200kg (Germany brand tecsis)
Load plate position adjustment
Height change test
High-precision servo measuring system
Test Standard: EN 1957
Application & Test Purpose
The test machine combines the loading device (roll) with a hardness measuring device. Thus, firmness test can be carried out according to ASTM F 1566 or EN 1957, without the need for a rearrangement of the specimen, which would falsify the measured values. It can also display height change by servo measuring system.
After simply loosening the front frame reinforcement the specimen can be inserted into a flexible test base plate. This makes the testing device very effective and space-saving.
Software allows for a flexible parameter entry, the execution of different testing procedures, a graphical representation of measured values, test evaluation, and logging. Then the pre-set number of cycles is reached, the test is ended automatically.
Test Standard: EN1725
Applications & Test Purpose
The machine increases mattress fatigue test function, it simulate human sitting on the edge of mattress, with 1000N force to test the fatigue of mattress for 5000 cyclic according to EN1725
A) Quality assurance items
The guarantee period with FOC within one year (not including nonexpendable & transport and travel fees)
1. During the guarantee period, Haida shall supply free maintenance or replacement for the damaged part(just for nonexpendable part) induced by non-human damage;
2. If any quality problems occurs within the guarantee period, and Haida must provide on-site service, the transport and travel expense shall be borne by customer;
3. If any big quality problems occurs out of the guarantee period, Haida will provide a maintain service, the transport and travel expense shall be borne by customer, also charge for a favorable price;
4. Haida will provide a lifetime favorable price to the buyer with the materials and spare parts used in system operation, equipment maintenance;
B) The main quality warranty maintenance certificate
If there is any dispute please refer to our guarantee letter, so:
① please keep the guarantee letter, if you lost it, please connect with us in a month.
② If the guarantee letter has been altered or it has no our stamp, it is useless.
C) The following conditions need to be paid reasonably even in the assurance period:
① Natural reasons
② Operating mistakes
③ Voltage is not fit for our operation instruction
④ Repack it without our guides
⑤ Damaged for borrowing to others
⑥ Damaged for authorized machine modification
⑦ Damaged for authorized calibration
⑧ Authorized transshipment mistake
⑨ Serve for long distance area
① Any service outside Guangdong, China, transport and travel fees shall be paid by customer.
② The apparatus try not to be used in the following situations:
a. Vibration, rocking the occasion.
b. Direct sunlight.
c. Hot, dusty, damp places.
d. To ensure safe, AC supply of the machine should be well grounded.
e. Do not use strong solvents (such as: benzene, nitro oil) washing machine.
f. Do not inject water and debris into the machine; prevent damage to electrical components and electrical shock.
g. Instrument displays the disassembly and debugging can only be measured by the State Department approved the units and the company, other people not allowed to overhaul.